What Is The Difference Between The ARTAS 9x And The Earlier Versions Of The Robot?May 26th, 2017
Q: What is the difference between the ARTAS 9x and the earlier versions of the robot? — T.J. ~ Washington, D.C.
A: The differences can be grouped into four broad categories:
1. Speed: The 9x is 20% faster than the 8x. This is achieved through the ARTAS robot’s ability to more quickly and accurately align with the follicles, faster movement from follicular unit to follicular unit while harvesting, and a shortened dissection cycle (less than 2 seconds). In addition, the 9x uses white LED lights instead of red, which permits an increased work flow from the ability to simultaneously incise and extractions grafts. The decreased strain on the eyes from the white lights (compared to red) makes this possible.
2. Accuracy: The 9x uses smaller needles that minimize wounding and donor scarring. It is especially useful for patients with fine hair or those who want to keep their hair short.
3. Functionality: The robotic arm on the 9x has a 1-inch base extender that gives the machine a longer reach and decreases the need for the patient to be repositioned. The ARTS 9x also has a smaller robotic head allowing the robot to harvest the grafts at a more acute angle. The ARTAS 9x also allows for more site making options due to the universal blade holder and the ability to program a change in the orientation of the incision in different regions of the scalp. The ARTAS 9x also uses a new harvesting halo to secure the tensioner (the grid-like device that indicates where the robot should harvest) which is faster to apply and more comfortable for the patient.
4. Use of Artificial Intelligence: The technology notifies the physician early-on if the harvesting is not precise, so that action can be taken to ensure most effective results. The ARTAS software can now detect areas with low (or no) hair density and prevent those areas from being over-harvested. This also decreases human error and saves time by automatically blocking these areas with low density. Finally, the ARTAS Hair Studio, can now create a 3-D image of the patient’s head with only one photo (as opposed to the prior requirement of 3 to 5).
Tags: ARTAS 9x, ARTAS Hair Studio, ARTAS Hair Transplant, ARTAS Robot Accuracy, ARTAS Robot Consistency, ARTAS Robot for FUE, ARTAS Robot Programmed to Avoid Scar, Restoration Robotics, Robotic Hair Transplant, Skin Tensioner
Posted by 2017-05-26 Updated 2018-01-26